Praxair Inc issued a recall of the Grab ‘n Go Vantage portable oxygen cylinder unit because fires may occur inside the cylinder unit causing the wall of the unit to break and release oxygen.
The fires are self-extinguishing and may be caused by physical impact such as dropping or knocking over the cylinder unit. If a fire occurs, users may experience burns or a lack of oxygen (hypoxia). This product may cause serious adverse health consequences, including death.
The Grab ‘n Go Vantage portable oxygen cylinder unit consists of an oxygen container and pressure regulator (valve). The pressure regulator (valve) is intended for medical purposes and is used to convert a medical gas pressure from a high variable pressure to a lower, more constant working pressure.
On November 26, 2012 Praxair Inc sent “Medical Device Correction” letters to customers and distributors describing the product, problem, and actions to take.
Read the medWatch safety alert, including a link to the recall notice by clicking here.